For mounting to Bourdon tube pressure gauges or transmitters, for connection to measuring points with internal thread. Designed for pressure measurements for which the diaphragm separators of Series 1080 cannot be employed. The flow of the medium can damage the tongue separator. This must be taken into consideration when determining the measuring point. The separator is inseparably fixed with the pressure measuring instrument to a hydraulic measuring unit, either directly, via a cooling element or via a capillary tube of stainless steel. Type of construction Tongue separator. Diaphragm with longitudinal starshaped profile, all parts made of stainless steel, welded. Connection Form A acc. to DIN 3852
- Temperature of medium Up to 80°C (1/2 hour up to 140°C),
- other temperatures limits (up to 300°C) on request.
- Hydraulic transmission fluid Glycerine, sweet almond oil, others on request.